CITI-Sense-Mob sensor testing

= Status = 1) earlier tests failed, no sensor signal

Potential reasons
 * no power supply, - measure with multimeter
 * to little power (effect) to fire up the sensor board - - don't know the power
 * communication fails

On-device monitoring
 * Tested the sensor with usb directly to pc and checked that there is some communication in a terminal program
 * Also, open it and recheck if the sim card is correctly in the slot.

2) second test partially success, no location information, show some measured values

Potential reasons:
 * static electricity

Done activities: (goal): change server address to the new one. (goal): force sensor to recheck the SIM card existence
 * Send SMS to sensor SIM card (40923931), content =srv.dunavnet.eu
 * Sensor discharging from static electricity.
 * Sensor running without SIM card then with SIM card in order

3) third test: Temporal success with location identification and measurements :

location identification failed potential reasons:
 * some wire bad connectivity.

Done activities:
 * recheck wires connectivity
 * static electricity discharging.
 * Partial testing for different parts of the sensor.

4) fourth test: Without air sensors board the sensor send approximately half of the expected number of sending

Done activities
 * Running of sensor without embedded air quality sensors board for four hours.

Result:
 * sensor send 114 reading pre 4 hour where sensor settings were configured to send sense read every 1 min.

(Notes): * The sensor needs 15 after turn it on to stabilize its sensing and to provide reliable data. * Air quality embedded sensors need to be exchanged with new one every 2 year.

caution: sensor used techs for air contents measurement based on the using of ACID (harmful chemical substance) which sometimes cause bad smell.

Needed for future testing:
 * list of used commands for sensor configuration in order to simplify and accelerate testing.
 * Mobile SIM card for sensors configuration.
 * Sensor manual (uploaded)

= Dunavnet link = Dunavnet |Device 3 online data

GUI Last measured:
 * http://citisense.dunavnet.eu/index.php?id=3
 * http://srv.dunavnet.eu/rephandler/ecobus/355255040040264
 * dunavnet - citisense!3

Raw data from WFS server
 * sensor 4 (.xml):

Link status

 * lab testing 20Dec2013, position: 59.9755117, 11.0535750 (NILU lab)

internal sensor measures: Air_Quality_bike_sensor_parameter